Apply to the test and result record of Mono-Si or Poly-Si solar cell pieces in Solar
The equipment is specially in the sorting of solar mono-crystalline silicon and the poly-crystalline silicon cell pieces, it is the first choice for the solar cell and solar cell module manufacturer.
1.Guarantee A gradeSpectrum,A grade non-uniformity of intensity,A grade Instability of radiation.
2.Pneumatic and buffering contact, front probe contact, back face contact.
3.4-line survey; 14-bit 4channel high-speed synchronous acquisition card
4.10 parameters display; meansurement parameter displays in tabulation and graphic way
5.Irradiance automatically compensating; temperature automatically compensating
6.Voice counts off and prompts; counting prompts of flash time
7.24 hours continuous working; flash working life of imported Xenon Lamp is 300000 times.
|Lamp Spectrum||IEC60904-9 and JISC8933 [Class A]|
|Light Intensity||100mW/cm²( continuous adjustable range 20~120mW/cm² )|
|Uniformity of Irradiation||±2% [Class A]||≤±3%|
|Instability of Irradiation||≤±2% [Class A]||≤±2%|
|Consistency of Testing||≤±0.5%||≤±1%|
|Pulse Width of Single Flash||10ms|
|Effective Testing Area/Watts||200mm×200mm / 0.1W~5W|
|Measuring Voltage||0V~0.8V(resolution 1mV)|
|Measuring Current||0mA~20A(resolution 1mA)|
|Tested Electric Performances||Isc,Voc,Pmax,Vm,Im,FF,EFF,Temp,Rs,Rsh|
Solar Equipment QA Solar Cell Tester on solar panel