REOO Expert of solar panel production line
1.1,Simulated light source
a,High-power pulsed xenon lamp.
b,Spectral range is AM1.5 which meet the IEC60904-9 requirements of spectral irradiance distribution
c,Light intensity 100mW/cm2(Adjustment range 20~120mW/cm2)
d,Irradiation uniformity ±2%.
e,Irradiation instability ≤±2%.
f,Xenon lamp life time 300,000(German imports)
g,Counting the number of features with Flash(Replacement of zero reset)
h,The light under the direction of the vertical lighting
a,Effective test area--200mm×200mm
b,Effective test range--0.1W~5W
c,Measuring voltage--0V~0.8V(Resolution 1mV)
d,Measuring current-- 0 mA~20A(Resolution 1mA) .
e,Temperature measurement--15°C~35°C(Resolution 0.1°C).
a,Test repeatability accuracy--±0.5%
1.5,Test parameters--Isc,Voc,Pmax,Vm,Im,FF,EFF,Temp ,Rs,Rsh
1.6,Test interval--3 seconds / piece
1.7,Data Acquisition--I-V curve with 5000 data collection points
1.8,Calibration test conditions--Self-correction device(Including manually enter the compensation parameters, automatic / manual temperature compensation, automatic compensation of light intensity).
a,Automatic test temperature
b,Temperature compensation and display
c,test the temperature of temperature correction
a,Test timeand graphical display parameters(I-V characteristic curve)
b,Test results have converted to the standard test conditions, the ability
c,Voice reported that the number of test results and voice prompts
a,Interface and can record the current work shows IV curves
b,All components can be a major test parameters specified file saved to your hard disk
High efficiency solar cell tester